Q: What output options does the GPC / GPCM Monitor offer
A: Output options offered are the following
> 4 to 20 MA
off of Pins 1 for MA (+ ) Pin 2 for MA ( - )
> Therm couple output
off of Pins 2 T/C ( - ) Pin 3 T/C ( + )
Thermocouple output is factory preset as inactive.
To adjust and set the type of thermocouple output, complete the following steps:
1. Press and hold the button, and press the button, then release them both. The Peak Hold display appears, see Figure 20.
2. Press the button several times until the thermocouple output indicator displays as shown in the following figure:
Press the ^ and I buttons to change the number on the display to the appropriate thermocouple output type. Use the following as a guide:
TC1 = J-type thermocouple output
TC2 = K-type thermocouple output
TC3 = E-type thermocouple output
TC4 = N-type thermocouple output
TC5 = R-type thermocouple output
TC6 = S-type thermocouple output
TC7 = T-type thermocouple output
4. Press the button until the temperature/value mode displays.
Note : The 4 - 20mA analog output is disabled when the thermocouple output is active.
Example picture from GPC / GPCM Pin outs